Slit-lamp structure mostly consists of two components, namely "slit-lamp" and "microscope". To be able to facilitate the fractured light irradiation around the parts on the eye from various angles, from distinctive angles and microscope eyes, needs on slit-lamp microscope in http://www.kanghuaa.com/handheld-portable-slit-lamp-23.html
machinery has adequate sway angle. Slit lamp light source requires its fractured edge need to be incredibly smooth, fracture have to be clear Imaging in left Center on the vertical plane, and focusing in the microscope will have to also focus on the circle around the vertical plane.
Slit lighting should have the following: 1. Slit width adjustable inside the variety of 0 to 14mm; 2. Crack length adjustable inside the variety of 1 to 14mm (when the aspect is 14mm cracks when the light is really a round spot); three. Fracture direction adjustable. Which suggests that cracks perpendicular to the light supply might be, may also be a horizontal, also can be inclined; 4.
adjustable brightness of your light supply;
For digital photo-slit lamp need to also have adjustable brightness from the background lighting.
Microscopes for stereoscopic binocular structure, it's essential to have: 1. Clear Imaging; 2. Show mirror with adjustable focal length to match the operator's different eye diopter; 3. Distance in the eyepiece adjustable to accommodate unique interpupillary distance of the operator;
Swing of mechanical structures furthermore to the above functions, also have 3 dimensions mobile Workbench with adjustable; device could be fastened to the patient's skull and jaw, jaw jaw holder is often adjusted up or down on the shelf to accommodate different patient's skull
length of strong lamp to prevent unconscious patient's eyes turned.